Research Article, J Spine Neurosurg Vol: 7 Issue: 4
Preliminary Clinical Outcomes From The Polyetheretherketone On Ceramic Simplify™ Disc FDA IDE Trial
Received: August 01, 2018 Accepted: August 08, 2018 Published: August 16, 2018
Citation: Maislin G, Maislin DG, Keenan BT, Alvis MR (2018) Preliminary Clinical Outcomes from the Polyetheretherketone on Ceramic Simplify™ Disc FDA IDE Trial. J Spine Neurosurg 7:4. doi: 10.4172/2325-9701.1000304
Background: This study was performed to evaluate the preliminary clinical results for the Simplify™ Cervical Artificial Disc. Methods: We compared outcomes for the first 61 subjects to reach Month 12 follow-up in a prospective, multicenter, FDA IDE clinical trial with 61 propensity score matched historical control subjects who received conventional anterior cervical discectomy and fusion (ACDF) for single-level cervical degenerative disc disease. The outcome measures included the change from preoperative baseline to Month 12 in Neck Disability Index (NDI) and visual analog scales (VAS) for neck and arm pain with missing follow-up determined by last observation carried forward. Results: The null hypothesis that the Simplify 1-disc is inferior to ACDF (non-inferiority margin=8.4) was rejected at a 1-sided p<0.0001. The upper bound of the 1-sided 95% non-inferiority confidence interval was -7.44 which is much smaller than the non-inferiority margin of 8.4. Superiority was demonstrated with a 2-sided p=0.0004. The upper bound of the 2-sided 95% confidence interval was -6.26 which is much less than zero. Sensitivity analyses on the assumptions about missing data and the matching included completer’s analyses and analyses that were restricted to the 55 of 61 first stage matches that could be achieved without expanding of the calipers used to identify potential matches. The p-values for non-inferiority in all analyses are <0.0001. Similarly, the p-values for superiority are all ≤ 0.0030. Conclusion: Therefore, we conclude that the Simplify Disc is superior to ACDF control in terms of improvement in NDI and VAS from baseline to Month 12.
Keywords: Cervical disc arthroplasty; PEEK (Polyetheretherketone)
Current FDA-approved total disc replacements (TDR) are typically comprised of metallic endplates (cobalt-chromiummolybdenum, titanium, or metal/ceramic composite), often with a polymer-based core comprised of ultrahigh molecular weight polyethylene or polyurethane. The clinical success of these discs attests to their safety, effectiveness, and durability [1,2], but postsurgical magnetic resonance (MR) imaging at the operative and adjacent levels of the cervical spine can be severely limited by the artifact induced by the current commercially available artificial discs with metallic endplates [3-5].
Magnetic resonance is the preferred mode for diagnostic imaging prior to spine surgery, but computed tomography (CT) use markedly increases after complex spine surgery involving implants . To minimize exposure to ionizing radiation  and concomitant risk of cancer , a cervical artificial disc that permits clear visualization of the operative and adjacent levels with MR would be preferable to current designs.
The use of PEEK for spinal implants continues to increase , primarily due to its mechanical properties and positive imaging properties (i.e. no metallic artifact on MRI or CT). The purpose of this study was to evaluate the preliminary results from the first subset of subjects to reach Month 12 follow-up in the single-level FDA Investigational Device Exemption (IDE) study for the PEEK-onceramic Simplify™ Artificial Cervical Disc.
Materials and Methods
Description of implant
The Simplify Disc is a three-piece intervertebral prosthesis consisting of two titanium-coated polyetheretherketone (PEEK) endplates and a mobile zirconia-toughened alumina (ZTA) core (Figure 1). The design of the Simplify Disc was based on the KineFlex™|C Cervical Artificial Disc (SpinalMotion, Mountain View, CA), which consisted of two cobalt-chromium-molybdenum (CCM), titanium-coated endplates with a biconvex CCM core. An IDE clinical trial  of the Kineflex|C Disc with five-year follow-up demonstrated excellent clinical results and validated the geometry of this three-part design with dual articulations.
In addition to providing motion and height restoration, the Simplify Disc is designed to permit subsequent visualization of cervical anatomy using MR imaging without the significant artifact. Along with changes in endplate and core materials, the Simplify Disc design was also optimized based on anatomical measurements taken during the Kineflex|C IDE.
The system is available in multiple configurations of footprint, height, lordosis and titanium coating thickness. All endplates feature smooth concave articulating surfaces to permit ±12° flexionextension and lateral bending, unlimited axial rotation, and a limited amount (<1.6 mm) of translation in the horizontal plane.
The Simplify Disc is provided packaged, preassembled and inserts as a single unit using a streamlined three-step procedure following a complete discectomy. A variation of the Simplify Disc (Kineflex Prime Disc) has been implanted in South Africa since 2013, and the Simplify Disc has been commercially available in the UK and Germany since 2016 .
Parent clinical study
The parent study is a prospective, controlled, multicenter clinical trial (US FDA IDE #G140154, NCT02667067) intended to demonstrate that the Simplify Disc is at least as safe and effective as conventional anterior cervical discectomy and fusion (ACDF) when used to treat one level between C3 to C7 for cervical degenerative disc disease (DDD) defined as intractable radiculopathy (arm pain and/or a neurological deficit) with or without neck pain or myelopathy due to a single-level abnormality localized to the level of the disc space in subjects who are unresponsive to conservative management.
The parent study is designed to utilize a non-concurrent historical control group with subject-level data in a parallel group design. The historical control group for both the parent study and for the current study will be formed from the randomized ACDF arm (N=133) of the completed multi-center, prospective, randomized clinical study of the Kineflex|C Disc trial  that compared the Kineflex|C Disc to conventional ACDF for treatment of subjects with single level degenerative disc disease (DDD) who are symptomatic at only one level from C3 to C7 that is unresponsive to conservative management. The first subject was treated on July 19, 2005 and the last randomized subject was treated on August 30, 2007. A total of 348 subjects were treated at 21 investigational sites in the United States, 192 subjects in the investigational Kineflex|C Disc treatment group (135 randomized and 57 non-randomized) and 134 subjects in the control group (133 randomized and 1 non-randomized) (all randomized in a 1:1 ratio). The parent study confirmed control group comparability and controlled for selection bias using propensity score subclassification  and prospectively enrolled 152 subjects from sixteen (16) sites between February 2016 and February 2018.
The parent study is utilizing a two-year composite clinical success (CCS) endpoint as the primary effectiveness endpoint. Individual success requires at least a 15-point improvement in the Neck Disability Score (NDI) at 24 months compared with baseline, maintenance or improvement in neurologic status at 24 months compared with baseline, no device failures or revision, reoperation, removal and/or supplemental fixation within 24 months of index procedure, and the absence of major adverse events within 24 months.
Statistical analysis plan for current study
The current study used an a priori design of a matched observational study including the use of propensity scores (PS). Only the subset of investigational subjects due for their Month 12 followup at the time of PS matching was eligible for inclusion. This provided N=61 investigational subjects. Nearest neighbor matching using Mahalanobis distance matching within propensity score “calipers”  was performed to select individual 1 to 1 matches for each investigational subject from the control group.
The primary effectiveness endpoint for the current study is the change from baseline to Month 12 post index surgery in the Neck Disability Score (NDI). For subjects experiencing a secondary surgical intervention (SSI), the last NDI score prior to the SSI was used to determine change scores.
The statistician developing the PS matched sets was blinded without access to outcome data. Balance between device groups was verified using a “Love plot”  that compares covariate balance between groups before and after the PS design.
The PS design of the observational study involved constructing a 1:1 Mahalanobis distance-matched sample of investigational device and ACDF control subjects. A caliper width of 0.25 pooled log(PS) standard deviations (on the natural log scale) was used . If there were investigational subjects with no ACDF control match within 0.25 pooled log(PS) standard deviations, the matching process was repeated for the unmatched investigational subjects with wider PS calipers for the initially non-matched investigational device subjects. Analyses were repeated including only first stage (higher quality) matches and then including both first stage and second stage matches.
The PS model was estimated using logistic regression that included important pairwise interactions among all covariates and the squares of continuous variables. The baseline variables listed in Table 1 were used in the construction of the matched pairs.
|Baseline Demographic Covariates||Baseline Covariates|
|Age||Average Disc Height*|
|Race (Caucasian vs non-Caucasian).||VAS Neck and Arm Pain|
|BMI||Any sensory deficit|
|Smoking Status||Motor mean < 5 (any abnormality)|
|At least 6 months of prior conservative treatment (Yes vs No)|
|Presence of progressive symptoms (Yes vs No)|
|Signs of nerve root compression (Yes vs No)|
Table 1: Baseline variables.
The primary effectiveness outcome was change from baseline to Month 12 in the Neck Disability Index (NDI). NDI has strong and welldocumented convergent and divergent validity with other instruments used in the evaluation of patients and subjects with neck pain”.
The minimum clinically important difference (MCID) in cervical spine fusion surgery was established to be 15 points out of 100 . When comparing effectiveness between two alternative treatments, the MCID is generally too large to serve as the non-inferiority margin. Therefore, the non-inferiority margin was a priori defined as the minimum detectable change (MDC), which was established for patients with mechanical neck pain as 8.4 .
The primary effectiveness hypothesis is that the Simplify Disc is not clinically inferior to the ACDF control in terms of mean difference in the NDI changes from baseline to 12 months.
The primary hypothesis was tested using paired t-test methods. Therefore, the null and alternative non-inferiority hypothesis may be represented as follows:
Ho: μ[Simplify-ACDF] ≤ δ
Where μ[Simplify-ACDF] is the mean within matched pair difference in changes from baseline to Month 12.
For this study, the non-inferiority margin was set to the MDC of 8.4. The primary non-inferiority test was conducted based on the upper bound of a 95% 1-sided confidence interval for the within match mean difference. If non-inferiority was demonstrated, superiority was tested by comparing this upper bound to zero using a 1-sided type 1 error rate of 0.025. If this upper bound was less than zero, then superiority of the Simplify disc relative to ACDF control was concluded.
Sample size analysis for non-inferiority was based on the paired t-test corresponding to the 1-sided 95% confidence interval. The standard deviation of changes from baseline to Month 12 in the ACDF controls was 20.7. If the correlation between changes from baseline to Month 12 between investigational device subjects and ACDF subjects is 0.50, then the SD of the paired differences is also 20.7. To be conservative, for purpose of sample size analysis, the SD of the paired differences was assumed to be equal to 22. Under this assumed SD of 22 and the one-sided significance level of 0.05 for a paired t-test examining non-inferiority, a sample size of 44 results in 80% power to reject the null hypothesis that the investigational and control are not equivalent at our a priori δ=8.4. Since the expected sample size was 61, the study had sufficient power to reject the null hypothesis of inferiority under the above assumptions.
If non-inferiority was demonstrated, superiority was tested at a 1-sided type 1 error rate of α=0.025. When the sample size is 56, a single group t-test with a 0.025 one-sided significance level has 80% power to detect the difference between a null hypothesis mean of 0.0 and an alternative mean of 8.4. Therefore, our sample of 61 patients results in sufficient power for superiority.
An NDI responder analysis was performed as a secondary endpoint. An improvement of at least 15 points from baseline to Month 12 defined a responder. A McNemar’s analysis of correlated proportions was used to compare these proportions between groups. A conditional logistic regression was used to determine a 95% confidence interval for the McNemar’s odds ratio comparing the likelihood of achieving at least a 15 point improvement among investigational device subjects relative to control subjects. Paired differences in changes from baseline in a visual analog scale (VAS) for neck and arm pain was similarly analyzed.
Results and Discussion
Results of PS matching analysis
Covariates included in the PS matching analysis are summarized among eligible Simplify Disc investigational and ACDF control patients in Table 2. There was generally good balance between the two arms as reflected in relatively small standardized mean differences (around 0.2 or smaller). However, a number of covariates have at least small differences (effect size (ES)>0.2) as defined by Cohen , including age (p=0.155, ES=-0.214), VAS (p=0.018, ES=0.376), no reported motor abnormality (p=0.018, ES=-0.350), at least 6 months of prior conservative treatment (p=0.062, ES=0.301), and presence of progressive symptoms (p=0.099, ES=0.261). Therefore, although only a few variables show statistically significant differences between study arms, better covariate balance for several variables can be achieved through the PS matching design.
|Measure||Simplify Disc||ACDF Control||P||Effect Size|
|Age, years||42.7 ± 8.7||44.4 ± 7.4||0.155||-0.214|
|BMI, kg/m2||28.0 ± 4.9||28.8 ± 5.6||0.371||-0.142|
|Current/Former Smoker, %||37.7%||47.4%||0.208||-0.196|
|NDI||62.8 ± 12.5||61.8 ± 13.0||0.633||0.074|
|VAS||81.0 ± 13.3||75.7 ± 14.8||0.018||0.376|
|Average Disc Height, mm||3.18 ± 0.73||3.27 ± 0.79||0.429||-0.124|
|No Sensory Deficit, %||50.8%||48.9%||0.801||0.039|
|No Motor Abnormality, %||37.7%||54.9%||0.026||-0.350|
|≥6 Wk Prior Conservative Treatment, %||83.6%||87.2%||0.500||-0.102|
|Progressive Symptoms, %||73.8%||61.7%||0.099||0.261|
|Nerve Root Compression, %||65.6%||64.7%||0.902||0.019|
Table 2: Baseline variables included in PS algorithm.
Variables for inclusion in the final PS model include all main effects, as well as important higher-order terms. Interactions between two categorical variables where one variable was smoking, sensory or motor deficits were excluded from the selection algorithm, as these led to unstable main effect estimates. The following variables were included in the final PS model.
• Average Disc Height
• No Sensory Deficit
• No Motor Abnormality
• ≥ 6 Weeks Prior Conservative Treatment
• Progressive Symptoms
• Nerve Root Compression
• (Age) × (NDI)
• (NDI) × (Smoking)
• (Sex) × (Progressive Symptoms)
• (BMI) × (Race)
In the first stage of PS matching, 55(90.2%) of the Simplify Disc investigational subjects were matched to corresponding ACDF control subjects. All standardized mean differences within the PS matched sample were ≤ 0.176 standard deviations in absolute value fell. This value is smaller than 0.20, the value typically associated with small differences as defined by Cohen  and generally reflected differences with little clinical significance. The standardized mean difference in the PS logits was only 0.078. Thus, the designed sample achieved with PS matching has substantially improved covariate balance. Therefore, the goals of PS matching were achieved in the first-stage design with included 55 (90.2%) of device participants with control matches falling within the specified caliper.
Not all investigational patients were matched to control participants in the first-stage of the PS design. There were six device patients without adequate matches in the first-stage. Unmatched patients tended to have higher propensity scores. The matching calipers were extended so that matches could be obtained for these patients and then added to the firststage designed sample, to achieve a second-stage design that retains all device participants. The resulting standardized differences and effect size percent reduction achieved in this second-stage design are illustrated and summarized in Figure 2 and Table 3, respectively.
|Measure||Before Match||After Match||Percent
|Logit of Propensity Score||1.135||1.032||0.200||0.182||82.4|
|Current/Former Smoker, %||-0.097||-0.196||0.049||0.100||49.0|
|Average Disc Height, mm||-0.095||-0.124||0.035||0.046||62.9|
|No Sensory Deficit, %||0.019||0.039||0.049||0.098||0.0|
|No Motor Abnormality, %||-0.172||-0.350||-0.033||-0.067||80.9|
|≥6 Wk Prior Conservative Tx, %||-0.036||-0.102||-0.049||-0.139||0.0|
|Progressive Symptoms, %||0.121||0.261||0.016||0.035||86.6|
|Nerve Root Compression, %||0.009||0.019||-0.016||-0.034||0.0|
**Calculated as the percent reduction in the absolute effect sizes before and after PS matching; values are presented as 0.0 for variables that demonstrated an increase, rather than reduction, in absolute effect size after PS matching design. In this study, increases were generally for variables with small differences in the full sample and did not result in clinically meaningful effect size differences
Table 3: Covariate differences in before and after PS matching sample.
The PS logit effect size increased from 0.078 to 0.182, yet all standardized mean differences still were all below 0.2. Thus, adequate covariate balance was achieved even when retaining all investigational participants. By including all investigational device subjects, external validity is maximized since the analysis set does not differ from the indicated population.
Primary effectiveness results for NDI
There was one subject in both groups experiencing a secondary surgical intervention. Clinical data after the SSI was censored and the values observed before the SSI were carried forward in the primary last observation carried forward (LOCF) analysis. The SSI’s occurred on follow-up days 118 and 168 in the investigational and control group, respectively. There was no clinical data post SSI for the control patient.
The mean within pair difference in changes from baseline to Month 12 in NDI was -13.41 (SD=27.93). The null hypothesis of inferiority is rejected t(60)=-6.10, p<0.0001.
The 90% confidence interval for the mean difference is (-19.39 to -7.44). Since -7.44 is smaller than 8.4, the null hypothesis of inferiority is rejected.s
Since non-inferiority was demonstrated, superiority was tested. The null hypothesis of equality is rejected, t(60)=-3.75, p=0.0004 and it is concluded that the Simplify device is superior to ACDF controls in terms of mean improvements from baseline to Month 12 in NDI.
The 95% confidence for the mean difference is (-20.56, -6.26). Since -6.26 is smaller than zero, it may be concluded that Simplify is superior to ACDF control in terms of mean change from baseline to Month 12 in NDI.
Three sensitivity analyses were conducted. These included:
Completers analysis based on all matches (N=46)
LOCF analysis based on Stage 1 matches (N=55)
Completers analysis based on Stage 1 matches (N=42)
Table 4 summarizes the results from the primary analyses and for these sensitivity analyses. In all cases the sensitivity analyses confirmed the results from the primary analyses, demonstrating both non-inferiority and superiority of the investigational device relative to control.
|N||Mean||SD||p-value*||90% LB||90% UB**||p-value***||95% LB||95% UB|
|NDI LOCF - All||61||-13.41||27.93||<0.0001||-19.39||-7.44||0.0004||-20.56||-6.26|
|NDI Completers – All||46||-10.26||22.19||<0.0001||-15.76||-4.77||0.0030||-16.85||-3.67|
|NDI LOCF - Stage 1 Only||55||-15.67||26.29||<0.0001||-21.61||-9.74||<0.0001||-22.78||-8.57|
|NDI Completers - Stage 1 Only||42||-11.24||21.48||<0.0001||-16.81||-5.66||0.0016||-17.93||-4.55|
**The upper bound of a two-sided 90% confidence interval (CI) is equivalent to the upper bound of a one-sided 95% CI for non-non-inferiority
***2-sided p-value for superiority
Table 4: Summary of primary effectiveness and sensitivity analysis paired differences in changes from baseline to Month 12 in NDI.
Among the 61 matched pairs, improvements of at least 15 points in NDI were observed for both members of the pair in 45 (73.8%) cases. There were no instances in which both members of the pair failed to achieve a 15 point improvement.
There were 16 discordant pairs, that is pairs in which one member achieved at least a 15 point improvement in NDI and the other did not. For 14 of 16 (87.5%) pairs, the Simplify patient achieved a 15-point improvement and the control patient did not. The McNemar’s odds ratio was equal to 7.0. That is, it was seven times more likely for patients treated with ACDF to fail to achieve a 15 point improvement compared to patients implanted with the Simplify disc.
A McNemar’s exact test was used to test the null hypothesis of quality in the likelihood of achieving a 15 point improvement among discordant pairs. This is equivalent to testing that the McNemar’s odds ratio is equal 1.0. The McNemar’s exact two-sided p-value is 0.0042 and, therefore, the null hypothesis of equality is rejected. The 95% confidence interval for the odds ratio is (1.59 to 30.5).
Change from baseline to Month 12 in VAS neck and arm pain was evaluated as a secondary endpoint. There was no a priori noninferiority margin specified, and so no formal test for non-inferiority was conducted. Nonetheless, the upper bound of a 1-sided 95% confidence interval may be used to determine the magnitude of differences that can be ruled out with “95% confidence”. The largest upper bound is 1.63. Therefore, by these analyses, it can be concluded that the mean improvement in VAS neck and arm pain for the Simplify Disc was no worse than 1.63 smaller than for ACDF controls. In fact, the mean difference was a least -5.46 across all analysis scenarios. Nominal superiority (two-sided p<0.05) was observed for the full analysis sample using LOCF (p=0.016) and when applying LOCF to the stage 1 matches only (p=0.007). There was one additional Simplify subject missing Month 12 VAS neck and arm pain that had non-missing NDI, reducing the Completers analysis set for VAS from 46 to 45.
The fact that all differences were negative and the largest 1-sided upper bound was only slightly positive supports the findings from primary analyses involving NDI (Table 5).
|N||Mean||SD||90% LB||90% UB*||p-value**||95% LB||95% UB|
|VAS LOCF - All||61||-11.41||35.76||-19.06||-3.76||0.0155||-20.57||-2.25|
|VAS Completers - All||45||-5.82||26.11||-12.36||0.72||0.1418||-13.67||2.02|
|VAS LOCF - Stage 1 Only||55||-13.36||35.32||-21.33||-5.39||0.0070||-22.91||-3.82|
|VAS Completers - Stage 1 Only||41||-5.46||26.96||-12.55||1.63||0.2019||-13.97||3.05|
**2-sided p-value for superiority
Table 5: Summary of secondary effectiveness and sensitivity analysis: Paired differences in changes from baseline to month 12 in VAS pain.
The results from this rigorously designed matched pairs analysis provide robust evidence that the Simplify Disc is not clinically inferior to ACDF control in terms of improvements in NDI from baseline to Month 12 (p<0.0001). Moreover, there was substantial evidence that the Simplify Disc is superior to ACDF in this regard (p ≤ 0.0030). Secondary analysis of VAS neck and arm pain support the finding that the Simplify Disc is not clinically inferior to ACDF. Data collection for the parent study will continue through 24-month follow-up.
We would like to thank David C. Hovda, Beth Neil, and Melissa K.C. Lui at Simplify Medical for their assistance with this manuscript.
- Chen C, Zhang X, Ma X (2017) Durability of cervical disc arthroplasties and its influence factors: A systematic review and a network meta-analysis. Medicine (Baltimore) 96: e5947.
- Coric D (2014) ISASS Policy Statement - Cervical Artificial Disc. Int J Spine Surg 8: 1.
- Sekhon LH, Duggal N, Lynch JJ, Haid RW, Heller JG, et al. (2007) Magnetic resonance imaging clarity of the Bryan, Prodisc-C, Prestige LP, and PCM cervical arthroplasty devices. Spine 32: 673-680.
- Sundseth J, Jacobsen EA, Kolstand F, Nygaard OP, Zwart JA, et al. (2013) Magnetic resonance imaging evaluation after implantation of a titanium cervical disc prosthesis: a comparison of 1.5 and 3 Tesla magnet strength. Eur Spine J 22: 2296-2302.
- Fayyazi AH, Taormina J, Svach D, Stein J, Ordway NR, et al. (2015) Assessment of Magnetic Resonance Imaging Artifact Following Cervical Total Disc Arthroplasty. Int J Spine Surg 9: 30.
- Patel VV, Andersson GB, Garfin SR, Resnick DL, Block JE, et al. (2017) Utilization of CT scanning associated with complex spine surgery. BMC Musculoskelet Disord 18: 52.
- Linet MS, Slovis TL, Miller DL, Kleinerman R, Lee C, et al. (2012) Cancer risks associated with external radiation from diagnostic imaging procedures. CA Cancer J Clin 62: 75-100.
- Shuryak I,Sachs RK, Brenner DJ (2010) Cancer risks after radiation exposure in middle age. J Natl Cancer Inst 102: 1628-1636.
- Kurtz SM (2012) Applications of Polyaryletheretherketone in Spinal Implants: Fusion and Motion Preservation, in PEEK Biomaterials Handbook 201-220.
- Coric D, Guyer RD, Nunley PD, Musante D, Carmody C, et al. (2018) Prospective, randomized multicenter study of cervical arthroplasty versus anterior cervical discectomy and fusion: 5-year results with a metal-on-metal artificial disc. J Neurosurg Spine 28: 252-261.
- Simplify Disc Instructions for Use (2017) Simplify Medical, USA.
- Rosenbaum PR, DB Rubin (1983) The central role of the propensity score in observational studies for causal effects. Biometrika 70: 41-55.
- Austin PC (2014) A comparison of 12 algorithms for matching on the propensity score. Stat Med 33: 1057-1069.
- Ahmed A, Husain A, Love TE, Gambassi G, Dell'Italia LJ, et al. (2006) Heart failure, chronic diuretic use, and increase in mortality and hospitalization: an observational study using propensity score methods. Eur Heart J 27: 1431-1439.
- Rosenbaum PR, DB Rubin (1985) Constructing a Control Group Using Multivariate Matched Sampling Methods That Incorporate the Propensity Score. Am Stat 39: 33-38.
- Vernon H (2008) The Neck Disability Index: state-of-the-art, 1991-2008. J Manipulative Physiol Ther 31: 491-502.
- Carreon LY, Glassman SD, Campbell MJ, Anderson PA (2010) Neck Disability Index, short form-36 physical component summary, and pain scales for neck and arm pain: the minimum clinically important difference and substantial clinical benefit after cervical spine fusion. Spine J 10: 469-474.
- Cleland JA, Childs JD, Whitman JM (2008) Psychometric properties of the Neck Disability Index and Numeric Pain Rating Scale in patients with mechanical neck pain. Arch Phys Med Rehabil 89: 69-74.
- Cohen J (1988) Statistical Power Analysis for the Behavioral Sciences, New York, USA.